Dental Ultrasonic Led Piezo Scaler With Function Of Scaling Endo And Periodontal
Adapter input:230V AC 50/60Hz 150mA. Main Unit input: 24VDC 1.3A.
Output tip Vibration Frequency:28kHz+-3KHz. Water pressure: 0.01Mpa to 0.5Mpa(0.1bar to 5 bar). Weight of Main Unit: 0.65kg. Weight of adapter: 0.8kg. Electric shock protection type: class 2. Degree of protection against electric shock: B. Degree of protection against harmful ingress of water: ordinary equipment(IPX0), Foot switch(IPX1). Degree of safety of application in the presence of flammable anesthetic mixture with air or with oxygen or nitrous oxide: Equipment not suitable for being used in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide. The conditions of working environment. Environmental temperature: +10C to +40C. Environmental humidity: 0 to 80%.
Turn the water control switch to the max based on symbol as shown. Insert the plug of the foot switch to its socket. Connect one end of the water pipe to the water entrance, lock the water pipe tightly with locknut, and connect the other end to the pure water source. Select a suitable scaling tip as you need and screw it on the handpiece tightly by the wrench.
Generally, hold the handpiece in the gesture as a pen in hand. In clinical scaling, dont use the point of tip to contact with the teeth vertically, so as not to damage teeth and tip. The normal frequency is extremely high. Under the normal working state of scaling tips, a light touch and a certain to-and-fro motion will eliminate the tartar without heating. Overexertion and long-time lingering are forbidden. Press OPTIONbutton, switch to periodontal function, then PERIO model, indicator lights. After switching to periodontal function, select the right tips, and use wrench to fix it an the handpiece. Adjust the frequency based on current status, it can be periodontal scaling after adjusting the right frequency. PressOPTIONbutton, switch to endo function, and endo indicator lights.
Insert endo files in the front hole of the endo chuck and use endo wrench to tighten the nut so as to clamp the files. Put the endo files into patients root canal slowly, start the foot switch can conduct ultrasonic root canal therapy. In clinical treatment, adjust the frequency according to actual situation. After finishing operation, keep the machine working for 20-30 seconds on water supply condition in order to clean the handpiece and tips. Uninstall the tip, sterilize handpiece, tips, endo chuck, endo files and wrench. Note: please dont pull out the handpiece and tips when the root switch is stepped on and the machine is producing ultrasonic vibration. The item "Dental ultrasonic LED piezo scaler with function of scaling endo and Periodontal" is in sale since Monday, February 26, 2018. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical, Lab & Dental Supplies\Other Medical, Lab & Dental Supplies". The seller is "zhjess3" and is located in , . This item can be shipped worldwide.
- Bundle Listing: Yes
- Brand: Bomaoer
- MPN: Does Not Apply
- Country/Region of Manufacture: China